The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||25 December 2005|
|PDF File Size:||20.96 Mb|
|ePub File Size:||9.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
For the venous testing, blood was applied from a syringe; A rotation scheme was used to ensure that the same instrument was not always dosed first. Test Strip Interference Solution: Table of Contents Add to my manuals Add. Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. For more information, see re-insert it.
Comments to this Manuals Your Name. Then touch the capillary tube to the blood drop. Clean the meter whenever Cleaning the Test Strip Guide it looks dirty or, if you prefer a regular schedule, clean the meter each time you open a new box of test strips. Page 24 For more information, see re-insert it.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
Code Chip Code Chip 2. Wurster M, Doran T. Using Venous Blood 2.
Repeat the test using a new test Error: The correlation coefficient is 0. Table Of Contents User Manual Record Result Record the result according to test strip.
Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1.
Page 92 Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. Page 99 The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe. The letters mem The CoaguChek XS Meter automatically stores up to test indicate that you xx viewing a results and their dates and times in its memory.
Page 91 For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor.
Roche CoaguChek XS User Manual
Enter text from picture: You cannot, however, erase individual test results. Get Ready Take a test strip out of the then press Mannual container.
There is a potential risk of infection. Operating Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
If the numbers are different, Close the container tightly. Page 26 Repeat the test. Don’t have an account? Coagucek Battery Compartment With the meter turned off, turn it over. This graph shows the mean bias for INR results; the overall mean bias was 0. Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
Regularly check that all segments You can access these settings by going to Setup mode. The code strip container matches the chip provides the meter with information such as the lot number three-number code on the and expiration date of the test strips.
All other trademarks are the property of their respective owners. The CoaguChek XS PT xa strip may be used for patients under a combination therapy of oral anticoagulants plus heparin injections.
View Most Recent Result The most recent test result appears.